In my last post on Oct 17th I introduced Agilent's new 53200A family of Universal Counter / Timers. In the post I gave a general overview of various features and specs that place the 53200A series as the top universal counters on the market. In this post I am going to go in more depth on the 20 ps single shot resolution (SSR) spec for the 53230A. What SSR resolution represents is how well the counter can resolve an event in time where an event is a threshold on an edge. 20 ps SSR is an industry leading timing spec. Any counter measurement consists of at least two events (except maybe totalizing). To calculate the SSR of two events measurement we use the root sum of squares (RSS) so for the 53230A the SSR for a two edge measurement would be:

Keep in mind this is the resolution for a single two event measurement, we can achieve even better resolution by averaging multiple measurements together to eliminate random noise. Of course this is at the cost of decreased measurement speed. Now SSR resolution is most often associated with time interval measurements, but every counter measurement basically comes down to timing so the better the SSR of a counter the more digits of resolution you get in a frequency measurement.

I am going to give a quick demo that calculates the SSR of the 53230A prototype sitting at my desk. The setup I use for the demo consists of the 53230A universal counter, 33522A function generator, two BNC cables, and a BNC tee. A continuous squarewave is first fed to channel 1 of the counter and then it passes through the other BNC cable to channel 2, as shown in the figure (sorry for the pic quality it was taken with my phone). Since the counter is measuring the same event (rising edge of squarewave) out of the function generator on both channels we can ignore the jitter on the signal from the function generator. Now we are not interested in the actual time interval measurement of the counter because we don't know the electrical length of the BNC cable between channels 1 and 2. What we are interested in is the standard deviation of the time interval measurement we get using the counter's statistics capability. As shown in the screen shot, we get a standard deviation of 15 ps (circled in red). If we assume all of the time interval measurements are within 3 standard deviations, then the max resolution we are seeing in this two event measurement is about 22.5 ps. Now to get the SSR of the 53230A at my desk we have to use RSS backwards on 22.5 ps. The answer is approximately 16 ps, which means the 53230A at my desk is well within the industry leading SSR spec of 20 ps!

Keep in mind this is the resolution for a single two event measurement, we can achieve even better resolution by averaging multiple measurements together to eliminate random noise. Of course this is at the cost of decreased measurement speed. Now SSR resolution is most often associated with time interval measurements, but every counter measurement basically comes down to timing so the better the SSR of a counter the more digits of resolution you get in a frequency measurement.

I am going to give a quick demo that calculates the SSR of the 53230A prototype sitting at my desk. The setup I use for the demo consists of the 53230A universal counter, 33522A function generator, two BNC cables, and a BNC tee. A continuous squarewave is first fed to channel 1 of the counter and then it passes through the other BNC cable to channel 2, as shown in the figure (sorry for the pic quality it was taken with my phone). Since the counter is measuring the same event (rising edge of squarewave) out of the function generator on both channels we can ignore the jitter on the signal from the function generator. Now we are not interested in the actual time interval measurement of the counter because we don't know the electrical length of the BNC cable between channels 1 and 2. What we are interested in is the standard deviation of the time interval measurement we get using the counter's statistics capability. As shown in the screen shot, we get a standard deviation of 15 ps (circled in red). If we assume all of the time interval measurements are within 3 standard deviations, then the max resolution we are seeing in this two event measurement is about 22.5 ps. Now to get the SSR of the 53230A at my desk we have to use RSS backwards on 22.5 ps. The answer is approximately 16 ps, which means the 53230A at my desk is well within the industry leading SSR spec of 20 ps!

شركة نقل عفش بالمدينة المنورة

ReplyDeleteشركة نقل عفش بالدمام

easteldmam

east-eldmam

شركة نقل عفش بالدمام

ReplyDeleteشركة نقل عفش

شركة نقل عفش

http://khairyayman.eklablog.com/

شركة نقل عفش بالمدينة المنورة

ReplyDeleteشركة نقل عفش بجدة

شركة نقل عفش بالرياض

شركة نقل عفش بالدمام